Bakon BK498 Wrist Strap Tester
ESD Wrist Strap Tester for ESD Wrist band is the first line of defense against electrostatic charge build up on the human body.
They are designed to carry away this charge as soon as it is generated.
At the same time the wrist strap must remain electrically safe for the user. Thus, it is important to test every wrist strap regularly.
The ESD wrist strap tester checks these parameters under actual operating conditions with the strap on the wrist.
In order to protect sensitive electronic components, the maximum wrist strap resistance should not be more than 10 meg ohms and footwear resistance should not be more than 35 Mega ohms.
On the other hand, the resistance should not be less than 0.75 mega ohm as protection to the wearer from dangerous levels of voltage and current flow.
It clearly indicates whether or not the wrist strap is safe to use. For checking surface resistivity of other surface we have Z203 Surface Resistivity Meter.
Features & Benefits –
- Power – 9V Dry Cell
- Grounding Wire – 2.5m
- External Dimensions – 115(W) x 120(H) x 47(D)mm
- Weight – 0.46kg
Test Results –
|R < 800KΩ||RED(Low)||OFF|
|800KΩ ≤ R ≤ 9MΩ||GREEN(Good)||ON|
|R > 9MΩ||RED(High)||OFF|
- Testing ESD Wrist Strap
- Testing ESD Footwears